non coring needle
Special non-coring puncture needles for implantable port-catheter systems, consisting of surgical steel with a special punch-free bevel, tubing, closure valve and Luer-Lock connector.
Via port/catheter system continous or intermittent administration of pharmaceuticals, infusions, blood transfusions, blood-samples, parenteral nutrition, pain therapy or as central venous access perioperatively and more
Different sizes and diameters with various lengths are available depending on indications and flow rate.
- no punch defects
- needle are no bending at the tip
- less painful due to lower skin resistance at puncture
- verbesserte Sicherheit für die Patienten
- Luer-Lock connector made of polycarbonate (PC) (taxolverträglich, Latex- und PVC-free)
straight with Luer-Lock
Art.Nr. Diameter/mm Length/mm Gauge Extras PZN SFN 0750 G 0,7 50 22 straight with Luer-Lock
00560673 SFN 0930 G 0,9 30 20 straight with Luer-Lock
01060286 SFN 0950 G 0,9 50 20 straight with Luer-Lock
03425332 SFN 1535 G 1,5 35 17 straight with Luer-Lock
(optional with tubing)
10216563 SFN 1535 B 1,5 35 17 straight with tubing 12363624 SFN 1835 G 1,8 35 15 straight with Luer-Lock
(optional with tubing)
10838997 SFN 1835 B 1,8 35 15 straight with tubing 12363630 DPK 2035 2,0 35 114 with mandrin + needle free adapter 02429581 Unit (VE) 25 Stück
Instruction for use
Non-coring puncture needles for implanted port/catheter systems needle with tubing and clamp for implanted port/catheter systems
- Special needle made of surgical steel with a special punch-free bevel (Trocar)
- Grip and fixation wings are foldable and made of polyurethane (PUR)
- Connective tubing made of polyurethane ( PUR)
- Luer-Lock connector made of polycarbonate ( PC)
- Clamp and- removable grip made of Polyoxmethylene ( POM )
- Continuous or intermittent administration of pharmaceuticals, infusions, blood transfusions, blood samples, parenteral nutrition etc.
- Pain therapy
- Central venous access perioperatively, CVP-monitoring
- Inflammation or infection at portal site
- Coagulation disorders
- Portal system leakage
WARNINGS AND PRECAUTIONS
Skin and soft tissue infections or infections of the entire portal system are possible if sterile techniques
are not used.
Warning: Puncture of the port should be performed in the supine position to prevent air embolism.
Negative pressure gradients develop in the superior vena cava in the sitting position or when the torso
is elevated. During puncture of the port, air can be sucked into the system via the needle. Therefore the
closure mechanism of the infusion tubing should be checked during puncture or when changing
syringes. When not in use, the tubing should be occluded with a screw stopper.
It is recommended that the needle be left in place no longer than 48 hours.
- Have patient in a supine position
- Carefully disinfect the skin
- The use of sterile gloves is recommended but not mandatory
- Remove the needle from the peel packaging
- Connect the Luer connector to syringe or infusion
- Remove air from the tubing by flushing with 0.9% NaCl
- Fold wings of the fixation plate upwards to be used as a grip for advancement of the needle
- Remove the needle protection sleeve
- Gently hold the port and fixation between two fingers
- Advance the needle quickly through the port membrane perpendicular to the base of the port (no danger of bending of the needle due to special bevel of the tip)
- Fix the needle with the tape
Warning: Needles are available in various lengths to ensure that a space remains between the skin and
the fixation plate. This is to prevent dislocation of the needle by movement of underlying tissues
such as the muscles.
a) Interruption of infusion, removal of infusion system while needle still in place
- If not in use for longer periods the system should be filled with physiologic saline
- Clamp tubing with closure mechanism
- Connect appropriately sized Luer Lock to the connection site
b) Removal of the needle
- After and between injection of pharmaceuticals the portal system should be flushed with physiologic 0.9% saline (possible pharmaceutical incompatibility).
- The system must be filled with heparinized saline after each use.
- The needle may be removed toward the end of saline injection under slight positive syringe pressure.
- The port chamber should be held in place below the skin level during needle withdrawal.
- The injection site is then covered by a sterile band-aid.
Important: The needle should be removed only when clamp is closed to prevent blood flow into the
system. Do not use syringes less than 10ml volume to prevent excessive pressures.
Warning: Blood deposits in the system generally indicate faulty handling, leakage, or a defective septum
inside the port.
If a thrombus in the port is suspected, the use of fibrinolytic agents such as streptokinase or urokinase
is possible. These recommendations require observation of the manufacturer's guidelines regarding
dosages and contraindications. In pain therapy via peridural or intrathecal catheters the use of a bacterial
filter is recommended. It is prohibited to flush these systems.
Maximal flow rates (aqueous solutions) of various SFN®-Needles of various diameters and lengths to
determine infusion rate:
18 gauge, ED 1.3 mm, ID 1.0 mm, puncture length 20 mm = 1000 ml in 11 minutes
19 gauge, ED 1.1 mm, ID 0.8 mm, puncture length 20 mm = 1000 ml in 20 minutes
20 gauge, ED 0.9 mm, ID 0.6 mm, puncture length 20 mm = 1000 ml in 39 minutes
22 gauge, ED 0.7 mm, ID 0.4 mm, puncture length 20 mm = 1000 ml in 80 minutes
Single use medical product. Non-resterilizable. The needle is sterile and non-pyrogenic as long as pakkaging
is undamaged. The packed products should be stored at temperatures ranging from 5-35 °C.
Avoid direct sunlight. Do not use if expiration date on packaging is exceeded.